## Tangent Law**
**

If a small bar magnet is suspended in two mutually perpendicular uniform magnetic fields B and $B_{H}$ such that it comes to rest making angle $\theta$ with direction of field $B _{ H }$ then $B = B _{ H }$ tan $\theta$ When a bar magnet is kept in two mutually perpendicular magnetic fields then each field will exert a couple on the magnet tending to set it parallel to itself. The torque due to $B_{H}$ ries to rotate the magnet in anticlockwise direction while torque due to B in clockwise direction.In equilibrium torque due to $B_{H}=$ torque due to B$mB _{ H } \times AN = mB \times AS$ so $B = B _{ H } \frac{ AN }{ AS }= B _{ H } \tan \theta$ ## Tangent Galvanometer**
**

It is an instrument used to measure very small currents. It is a galvanometer with a fixed coil and moving magnet. It is based on tangents law.### Construction & Working **
**

It consists of a circular coil of a large number of turns of insulated copper wire wound on vertical circular frame of non magnetic material. This frame can be rotated about its vertical diameter as axis. A small magnetic compass needle pivoted at centre is free to rotate in horizontal plane over a horizontal circular scale graduated in degrees. The position of needle is read by a long aluminium pointer attached at right angles to magnetic needle.Initially coil is made vertical by levelling screws and coil is brought in magnetic meridian. Now $B_{H}$ is in plane of coil. As current is passed through coil magnetic field B is produced which is perpendicular to plane of coil.According to Tangent law $B=B_{H} \tan \theta$ or $\frac{\mu_{0} nI }{2 a }= B _{ H } \tan \theta$Or $I =\frac{2 a }{ n \mu_{0}} B _{ H } \tan \theta=\frac{ B _{ H }}{ G _{1}} \tan \theta= K \tan \theta$ where $G _{1}=\frac{ n \mu_{0}}{2 a }$ is galvanometer constant and $K =\frac{ B _{ H }}{ G _{1}}$$K =\frac{ B _{ H }}{ G _{1}}$If $\theta=45^{\circ}$ then = K, so reduction factor of a tangent galvanometer is numerically equal to the current in ampere through the galvanometer which produces a deflection of $45^{\circ}$ when the plane of its coil lies in the magnetic meridian. K depends on geometry of coil and value of $B_{H}$. #### Sensitivity

A tangent galvanometer is both sensitive and accurate if the change in its deflection is large for a given fractional change in current.$I=K \tan \theta \quad$ or$dI = K \sec ^{2} \theta d \theta$ or$\frac{ d I }{ I }=\frac{ d \theta}{\sin \theta \cos \theta}=\frac{2 d \theta}{\sin 2 \theta} \quad$ or $\quad d \theta=\frac{\sin 2 \theta}{2} \frac{ d I }{ I }$$d \theta=( d \theta)_{\max }$ if $\sin 2 \theta=1=\sin \frac{\pi}{2}$$SO$$\theta=\frac{\pi}{4}$The tangent galvanometer has maximum sensitivity when $\theta=45^{\circ}$.**Ex. Two tangent galvanometers differ only in the matter of number of turns in the coil. On passing current through the two joined in series, the first shows a deflection of $35^{\circ}$ and the other shows a $45^{\circ}$ deflection. Calculate the ratio of their number of turns. Use tan $35^{\circ}=0.7$**

**Sol.**In tangent galvanometer $\frac{\mu_{0} NI }{2 r }= B _{ H } \tan \theta$Here $N \propto \tan \theta$ so $\frac{N_{1}}{N_{2}}=\frac{\tan \theta_{1}}{\tan \theta_{2}}=\frac{\tan 35}{\tan 45}=\tan 35=\frac{7}{10}$Required ratio is 7 : 10

**Ex. Two tangent galvanometers A and B have their number of turns in the ratio 1 : 3 and diameters in the ratio 1 : 2 (a) which galvanometer has greater reduction factor (b) which galvanometer shows greater deflection when both are connected in series to a dc source?**

**Sol.**(a) $K =\frac{2 a }{ n \mu_{0}} B _{ H }$ so $\frac{ K _{1}}{ K _{2}}=\frac{ a _{1}}{ a _{2}} \cdot \frac{ n _{2}}{ n _{1}}=3 \times \frac{1}{2}=\frac{3}{2}$ or $K _{1}=\frac{3}{2} K _{2}$ so $K _{1}> K _{2}$ Galvanometer A has greater reduction factor(b) $\frac{\mu_{0} nI }{2 a }= B _{ H } \tan \theta$ so $\tan \theta \propto \frac{n}{a}$$\frac{\tan \theta_{1}}{\tan \theta_{2}}=\frac{n_{1}}{a_{1}} \times \frac{a_{2}}{n_{2}}=\frac{1}{3} \times \frac{2}{1}=\frac{2}{3}$so $\tan \theta_{1}>\tan \theta_{2}$ or $\theta_{2}>\theta_{1}$so B shows greater deflection.

**Also Read:**Biot Savart’s Law

**Click here for the Video tutorials of Magnetic Effect of Current Class 12**

For free video lectures and complete study material, DownloadAbout eSaralAt eSaral we are offering a complete platform for IIT-JEE & NEET preparation. The main mission behind eSaral is to provide education to each and every student in India by eliminating the Geographic and Economic factors, as a nation’s progress and development depends on the availability of quality education to each and every one. With the blend of education & technology, eSaral team made the learning personalized & adaptive for everyone.

**eSaral APP.**

Thanks a lot, quality content right there.

Thanks. Always happy to help. 😊

For more FREE resources for your exams, download the eSaral App.

Well explained.Students can follow the basic principle and different fectors effecting acting on the galvanometer easily.

it is very useful